
Order by:
Available to:

Buy It Now
$999.99 Shipping Condition: New – Open box Location: Ukraine High precision capacitance AC bridge P5026(Set) Multivalued. Highfrequency decade capacitance box capacitor standard. AC bridge analog of General Radio(GR. ... moreGenRad) IET Labs, but has the same or better characteristics. AC bridge Р5026М is designed to measure capacitance and dielectric tangent loss of highvoltage industrial insulation for a"direct" both outputs of the measuring object isolated from the"earth" and" inverted" one output of the measured object is connected to"earth" schemes under operational conditions. Directly on the installation location, as well as measurements of"direct" scheme in the laboratory capacity and dielectric tangent loss of various insulating materials and condensers at 50 and 60Hz. P5026 Set includes: 1. P5026 AC bridge 1psc. 2. Q5122 potential protector 1set. 3. P5069 condenser block 1psc. 4. P5023 Standart air condenser 1 set. 5. Solid dielectrics testing electrodes 1 set. 6. Power cord 1psc. 7. Cables 1 psc. 8. Manual and docs. Technical characteristics: M easurement range С х tgδ х 10 pF1000 pF 1∙1040.1 100 pF10000 pF 1∙1041.0 10 4 pF2∙10 4 pF 2∙10 4 pF10 5 pF 10 5 pF10 6 pF 5∙1041.0 100 pF10000 pF 5∙1031.0(10 4 pF10 5 pF)10 5 pF10 6 pF) 650 pF2∙10 5 pF 5∙1030.1 6500 pF2∙10 6 pF 5∙1031.0 2∙10 6 pF5∙10 8 pF Measurement range Operating voltage Measurement schem e С х tgδ х 10 pF1000 pF 1∙1040.1 5 kV10 kV« straight» with F 5122 and Р5069 100 pF10000 pF 1∙1041.0 3 kV10 kV 10 4 pF2∙10 4 pF 2∙10 4 pF 10 5 п pF 3 kV5 kV« straight» with F 5122 10 5 pF10 6 pF 5∙1041.0 100 pF10000 pF 5∙103 1.0 3 kV10 kV« straight» with« turned back» 10 4 pF10 5 pF)10 5 pF10 6 pF) 650 pF 2∙10 5 pF 5∙103 0.1 0.1 kV« straight» 6500 pF2∙10 6 pF 5∙1031.0 2∙10 6 pF5∙10 8 pF MTBF not less than: 5250 hours; The average service at least: 6 years; Dimensions: 540 x 280 x 380mm; Weight: 22kg. Warranty 12 month. AC Bridge Р5026М consists of the following components
Buy It Now
$90 Shipping Condition: New – Open box Location: Ukraine
Buy It Now
$175 Shipping Condition: New – Open box Location: Ukraine P5083 Multifunctional. Multivalued decade resistance box. Bridge, resistor. Potentiometer analog of: General Radio(GenRad) LEEDS& NORTHRUP, ESI, IET ... moreLabs, GR, Yokogawa, Shallcross, Tinsley& Co, Shallcross, Leeds& Northrup, L&N, Extech, EMCO, AEMC devices. But has the same or better characteristics: AC Bridge P5083 is an autonomous measuring device for general industrial purposes and designed for automatic measurements of: capacitance C. Inductance L. Active resistance R. Tangent loss tgδ( Q Qr) tangent angle of phase shift tgφ( Q Qc and QL) of measuremed objects. And also a percentage deviation of measured object with further presentation of measurement results in digital form. AC bridge P5083 can be used for: Scientific researches; Metrological measuring instruments attestation; Quality control of electrical and radiotechnical products; Nonelectrical values measurement with a use of nonelectrical values transformers into one of measured values. Min. Value of accuracy class: With an external model of calibration) 0.02/0,002(0.005/0,0002) The range of measured values : resistance. Ohm from 1·105 to 1·10 10 capacitance from 1·1016 to 10 inductance from 1·1010 to 1·10 7 angle shift tangent(tg φ) from ± 1·105 to ± 1·10 +5 deviation percentage. from ± 0.001 to ± 100 frequency.KHz(3000 discrete frequencies installed) from 0.1 to 100 Discrete digital frame(on low range of measurements) active resistance. Ohm 1·105 capacitance 1·1016 inductance 1·1010 AC range of values. Onsite measurements(dependingon the measuredvalues) V. From 0.01 to 30 Dimensions: 440 x158x405mm; Weight: 16 kg. Warranty 12mounth. Technical characteristics of AC bridge P5083: D epending on frequency and voltage of measurement object. Values measured by the AC bridge correspond to those shown in Table 1; Table1 Main Additional main derivative C Cp Rp tgδ c(Q R) Cs Rs L Lp Rp tgδ L(Q R) Ls Rs R Rp Cp tgδ c(Q C) Lp tgδ L(Q L) Rs Cs tgδ c(Q C) Ls tgδ L(Q L)
Buy It Now
$170 Shipping Condition: Used Location: Ukraine P593 Multivalued decade inductance box. Inductor standard set is analog of General Radio(GR. GenRad) IET Labs, but has the same or better characteristics. ... moreInductance standard P593 d esigned for use as an AC element in measuring systems with frequency of 1 kHz to 10000kHz. Standard’s nominal value of inductance 0.01 mkH 100mH Accuracy class: 1; 2; 5 Valued Inductance standard P593(hereinafter standard) are designed for use as an AC element in measuring systems with frequency of 1 kHz to 10000kHz. Inductance standard P593 designed for use in moderate climate conditions in dry heated rooms at ambient temperature of 10 to 35 ° C and relative humidity up to 80% at 25 ° C. P593 a set of: 15 standard inductors; Nominal value from 0.01mkH to 100mH; Accuracy class: 1. 2, 5; Accuracy of the attestation: from 0.2; Full average lifespan at least 10 years; Dimensions: 100x150mm for more 1mkH; 40x160mm for less 0.3 mkH; 490x335x185mm whole case; Weight: 1 standart 0.7kg; Whole set 15kg; Warranty 12mounth. Technical characteristic of Decade inductance box inductor standard set type P593: Nominal Value of inductance Accuracy class Normal Frequency KHz Max Current A Max Voltage V Q Q/Frequency KHz 100 mH 1 1 0.010 5.000 70 30 30 mH 1 1 0.010 1.000 40 30 10 mH 1 1 0.010 0.500 130 100 3 mH 1 1 0.010 0.100 160 100 1 mH 1 1 0.010 0.050 200 300 300 mkH 1 1 0.010 0.020 140 300 100 mkH 1 1 0.010 0.005 200 1000 30 mkH 1 1 0.250 0.050 130 1000 10 mkH 1 1 0.250 0.020 210 3000 3 mkH 1 100 0.250 0.500 180 3000 1 mkH 2 100 0.250 0.100 200 10000 0.3 mkH 2 100 0.500 0.050 0.1 mkH 2 100 0.500 0.020 0.03 mkH 5 100 0.500 0.005 0.01 mkH 5 100 0.500 0.002 Technical characteristics Inductance standard P593 correspond to GOST 21175. Inductance standard nominal value(L) accuracy classes(K) Normal frequency(f) attestation fixed frequencies. Maximum current values(I) maximum operating voltage values(U) lowest values quality factor(Q) for frequencies f shown in Table 1. Acceptable basic error limit. Sh
Buy It Now
$190 Shipping Condition: Used Location: Ukraine P593 Multivalued decade inductance box. Inductor standard set is analog of General Radio(GR. GenRad) IET Labs, but has the same or better characteristics. ... moreInductance standard P593 d esigned for use as an AC element in measuring systems with frequency of 1 kHz to 10000kHz. Standard’s nominal value of inductance 0.01 mkH 100mH Accuracy class: 1; 2; 5 Valued Inductance standard P593(hereinafter standard) are designed for use as an AC element in measuring systems with frequency of 1 kHz to 10000kHz. Inductance standard P593 designed for use in moderate climate conditions in dry heated rooms at ambient temperature of 10 to 35 ° C and relative humidity up to 80% at 25 ° C. P593 a set of: 15 standard inductors; Nominal value from 0.01mkH to 100mH; Accuracy class: 1. 2, 5; Accuracy of the attestation: from 0.2; Full average lifespan at least 10 years; Dimensions: 100x150mm for more 1mkH; 40x160mm for less 0.3 mkH; 490x335x185mm whole case; Weight: 1 standart 0.7kg; Whole set 15kg; Warranty 12mounth. Technical characteristic of Decade inductance box inductor standard set type P593: Nominal Value of inductance Accuracy class Normal Frequency KHz Max Current A Max Voltage V Q Q/Frequency KHz 100 mH 1 1 0.010 5.000 70 30 30 mH 1 1 0.010 1.000 40 30 10 mH 1 1 0.010 0.500 130 100 3 mH 1 1 0.010 0.100 160 100 1 mH 1 1 0.010 0.050 200 300 300 mkH 1 1 0.010 0.020 140 300 100 mkH 1 1 0.010 0.005 200 1000 30 mkH 1 1 0.250 0.050 130 1000 10 mkH 1 1 0.250 0.020 210 3000 3 mkH 1 100 0.250 0.500 180 3000 1 mkH 2 100 0.250 0.100 200 10000 0.3 mkH 2 100 0.500 0.050 0.1 mkH 2 100 0.500 0.020 0.03 mkH 5 100 0.500 0.005 0.01 mkH 5 100 0.500 0.002 Technical characteristics Inductance standard P593 correspond to GOST 21175. Inductance standard nominal value(L) accuracy classes(K) Normal frequency(f) attestation fixed frequencies. Maximum current values(I) maximum operating voltage values(U) lowest values quality factor(Q) for frequencies f shown in Table 1. Acceptable basic error limit. Sh
Buy It Now
$70 Shipping Condition: New – Open box Location: Ukraine Inductance standard P547. Analog of Lutron. Advantest. Avtech, HP Agilent. NoiseCom, General Radio. Boonton, Anritsu, Fluke and general Electric, but ... morehas the same or better characteristics. This device can be used for wide measurement range. For sale is 1 of 4 devices. Please choose on from the table. Is designed to be used as a inductance standard in AC circuits with a maximum operating frequency 1500 10000 Hz. Depending on inductance standard nominal value. Nominal value: 1 H. 0.1 H. 0.001 H. 0.0001 H Accuracy: 0.1% Condition: NEW(not used. An old stock) Tested and 100% workable. If necessary. We can provide calibration certificate from Government standardization institute(+145265usd) Also we can provide other resistance. Capacitance and inductance standards, decade box and other etalons. Also we can provide other resistance. Capacitance and inductance standards, decade box and other etalons. If you haven’t found the needed item in my listing. Please contact me. I’ll try to help you. Technical characteristics of device inductance standard P547: Inductance standard P547 is designed to be used as inductance standard in AC circuits with a maximum operating frequency 1500 10000 Hz, depending on inductance standard nominal value. Inductance standard P547 is produced with the following nominal values: 1 H. 0.1 H. 0.001 H. 0.0001 H. Inductor coil P547 is designed for operation at temperatures of 10 ° C to +35 ° C and relative humidity up to 80% Technical characteristics Parameters Parameters and characteristics value Note Deviation from nominal value a) ± 0.1% for all inductance values and in frequency range of not more than: L, H 1 0. 1 0. 01 0. 001 and 0. 0001 Hz max. 500 1500 5000 10000 b) inductance change at higher frequency relative to maximum operating frequency for"a" does not exceed 0.2% at L, H 1 0. 1 0. 01 Hz max. 1000 3000 10000 Ohm resistance do not exceed L H 1 0. 1 0. 01 0. 001 0. 0001 On standards indicated resistance at t = +20 ° C r ~ Ohm 140 20 1. 5 0
Buy It Now
$999.99 Shipping Condition: New – Open box Location: Ukraine
Buy It Now
$140 Shipping Condition: New – Open box Location: Ukraine P5017 Multivalued decade of mutual inductance. Inductance box, inductor standard set is analog of General Radio(GR, GenRad) IET Labs, but has the same ... moreor better characteristics: Variation range. MH + 12.99 Acceptable basic error limit. MH +(14х103 + 1.1х103 M) Inductance variation range From 4.5 to 10 Acceptable basic error limit. Deg + 0.4( M 0.1 mH) Variation range of residual mutual inductance. MH + 150 Accuracy class. MH + 15 C omplex resistance m odul of the primary circuit. Ohm 90 + 5 C omplex resistance a rgument of the primary circuit 0.45 + 0.02 and 0.55 + 0.02 C omplex resistance m odul of the secondary circuit. Ohm 400 + 20 C omplex resistance a rgument of the secondary circuit 0.1 Nominal Current. A 0.125 Normal frequency. Hz 50 MTBF: not less than 24500 Hours; Dimensions: 405x215x235mm Weight: 11kg; Warranty 12 mounths This device was also used in USSR and Russia army. So it has best quality. Longevity and permanence characteristic. Condition: NEW(not used. An old stock) Tested and 100% workable. If necessary. We can provide calibration certificate from Government standardization institute(+145265usd) Also we can provide other resistance. Capacitance and inductance standards, decade box and other etalons. Also we can provide other resistance. Capacitance and inductance standards, decade box and other etalons. If you haven’t found the needed item in my listing. Please contact me. I’ll try to help you. Complex mutual inductance box Р5017 is designed to control interchangeable differential transformer devices. Powered from AC frequency of 50 Hz. Measuring range of complex mutual inductance modulus M in range 12.99 mH. Measuring range of complex mutual inductance argument from 4.5 to 10 if M is not less than 0.1 mH. Measuring range of complex residual mutual inductance modulus M 0 within ± 150mkH. Complex mutual inductance box Р5017 is designed to control interchangeable differential transformer devices. Powered from AC frequency of 50 Hz. Complex mutual inductance box Р5017 is us


P331 Coil. Unevalued resistance standard resistor. USED. P331 resistance standard resistor analog of LEEDS& NORTHRUP. ESI, Honeywell, IET Labs device has the same or better characteristics: Nominal values 100 Ohm. For sale is one coil) I sold a more then 70 units of this devices types. That is why now in stock I had units with scratches. Cracks and cleavage of the case and panel. Please take a look the picture. Also I had no enough units with class 0.01. That is why I can send you unit with class 0.02. I will choose the best one used device I had in my stock. It will be tested and 100% workable. I give you a warranty for 1 year. I Also I had new devices please take a look them by the link: http:stores.ebay.com/electroradiodevicehighprecision/ResistanceStandardsBoxesi.html?fsub=3003745013. Accuracy class: 0.01% It is max error(accuracy) ... morewhich is given per manufacturer. It’s including nominal of the value. Temperature resistance change, Stability per year, stability per max current include and all other parameters which can affect the accuracy of the resistance standard) Real(typical) error: 0.002% = 20ppm; Accuracy of attestation(calibration uncertainty) 0.001% General characteristics(given by manufactur) Nominal Value Ohm Model Adjustment to Nominal. ppm) Stability per year. ppm) Resistance Change 1525°C from 20°C, ppm/°C) Power Coef(ppm/mW) Max. Current. A Insulation resistance. Mohm max typical max typical max typical max typical 100 P331 100 8 20 2 20 10 0.1 0.025 0.1 1000 Nominal power: 0.1 Watt; Max power: 1 Watt; Dielectric strength of insulation: 2kV; Height: 170mm; Diameter: 110mm; Weight: 1kg; Hole for thermometer. Operation temperature: 1525 °C; Relative humidity: 25 to 80% Atmosphere pressure: 84106.7 kPa(630800 mmHg) Warranty: 12 month. Electrical resistance coil P331 Represents an exemplary resistance standard and is designed for testing and fitting working resistance coils in DC circuits. As well as exemplary and working( laboratory) measuring devices. Nominal


P321 Coil. Unevalued resistance standard resistor. NEW. P321 resistance standard resistor analog of LEEDS& NORTHRUP. ESI, Honeywell, IET Labs device has the same or better characteristics: Nominal values 0.1 Ohm. For sale is one coil) Accuracy class: 0.01% It is max error(accuracy) which is given per manufacturer. It’s including nominal of the value. Temperature resistance change, Stability per year, stability per max current include and all other parameters which can affect the accuracy of the resistance standard) Real(typical) error: 0.002% = 20ppm; Accuracy of attestation(calibration uncertainty) 0.001% General characteristics(given by manufactur) Nominal Value Ohm Model Adjustment to Nominal. ppm) Stability per year. ppm) Resistance Change 1525°C from 20°C, ppm/°C) Power Coef(ppm/mW) Max. Current. A Insulation resistance. ... moreMohm max typical max typical max typical max typical 0.1 P321 100 15 20 3 20 4 0.1 0.05 3.2 1000 Nominal power: 0.1 Watt; Max power: 1 Watt; Dielectric strength of insulation: 2kV; Height: 170mm; Diameter: 110mm; Weight: 1kg; Hole for thermometer. Operation temperature: 1525 °C; Relative humidity: 25 to 80% Atmosphere pressure: 84106.7 kPa(630800 mmHg) Warranty: 12 month. Electrical resistance coil P321 Represents an exemplary resistance standard and is designed for testing and fitting working resistance coils in DC circuits. As well as exemplary and working( laboratory) measuring devices. Nominal resistance 0.1 Ohm; Accuracy class 0.01 Electrical resistance coil P321 represents an exemplary resistance standard and is designed for testing and fitting working resistance coils in DC circuits. As well as exemplary and working( laboratory) measuring devices. Electrical resistance coil P321 operates in liquid environment( capacitor oil, kerosene, o rganosilicium fluid) Acceptable resistance deviation of coils P321 from their nominal values caused by adjustment inaccuracy at temperature of 20°C and power not exceeding nominal: for accuracy class 0.01 is not more th


P321 or P331 Coil. Unevalued resistance standard resistor. P321 or P331 resistance standard resistor analog of LEEDS& NORTHRUP. ESI, Honeywell, IET Labs device has the same or better characteristics: Nominal values 1 or 10 or 100 or 1000 or 10000 or 100000 Ohm. For sale is one of coils. You may choose a value for your consideration) I sold a more then 70 units of this devices types. That is why now in stock I had units with scratches. Cracks and cleavage of the case and panel. Please take a look the picture. Also I had no enough units with class 0.01. That is why I can send you unit with class 0.02. I will choose the best one used device I had in my stock. It will be tested and 100% workable. I give you a warranty for 1 year. Accuracy class: 0.01% It is max error(accuracy) which is given per manufacturer. It’s including nominal of the ... morevalue. Temperature resistance change, Stability per year, stability per max current include and all other parameters which can affect the accuracy of the resistance standard) Real(typical) error: 0.002% = 20ppm; Accuracy of attestation(calibration uncertainty) 0.001% General characteristics(given by manufactur) Nominal Value Ohm Model Adjustment to Nominal. ppm) Stability per year. ppm) Resistance Change 1525°C from 20°C, ppm/°C) Power Coef(ppm/mW) Max. Current. A Insulation resistance. Mohm max typical max typical max typical max typical 1 P321 100 15 20 3 20 4 0.1 0.05 1 1000 10 P321 100 15 20 3 20 4 0.1 0.05 0.32 1000 100 P331 100 8 20 2 20 10 0.1 0.025 0.1 1000 1000 P331 100 8 20 2 20 10 0.1 0.025 0.032 1000 10000 P331 100 8 20 2 20 10 0.1 0.025 0.01 5000 100000 P331 100 8 20 2 20 10 0.1 0.025 0.0032 50000 Nominal power: 0.1 Watt; Max power: 1 Watt; Dielectric strength of insulation: 2kV; Height: 170mm; Diameter: 110mm; Weight: 1kg; Hole for thermometer. Operation temperature: 1525 °C; Relative humidity: 25 to 80% Atmosphere pressure: 84106.7 kPa(630800 mmHg) This device was also used in USSR and Russian army. So it has best quality. Longevity. And pe


P310 Coil. Unevalued resistance standard resistor. P310 resistance standard resistor analog of LEEDS& NORTHRUP. ESI, Honeywell, IET Labs device has the same or better characteristics: Nominal values 0.001 or 0.01( For sale is one of coils. You may choose a value for your consideration) I sold a more then 70 units of this devices types. That is why now in stock I had units with scratches. Cracks and cleavage of the case and panel. Please take a look the picture. Also I had no enough units with class 0.01. That is why I can send you unit with class 0.02. I will choose the best one used device I had in my stock. It will be tested and 100% workable. I give you a warranty for 1 year. Accuracy class: 0.01% It is max error(accuracy) which is given per manufacturer. It’s including nominal of the value. Temperature resistance change, Stability ... moreper year, stability per max current include and all other parameters which can affect the accuracy of the resistance standard) Real(typical) error: 0.002% = 20ppm; Accuracy of attestation(calibration uncertainty) 0.001% General characteristics(given by manufactur) Nominal Value Ohm Model Adjustment to Nominal. ppm) Stability per year. ppm) Resistance Change 1525°C from 20°C, ppm/°C) Power Coef(ppm/mW) Max. Current. A Insulation resistance. Mohm max typical max typical max typical max typical 0.001 P310 100 20 20 5 20 5 0.1 0.05 32 1000 0.01 P310 100 20 20 5 20 5 0.1 0.05 10 1000 Nominal power: 0.1 Watt; Max power: 1 Watt; Dielectric strength of insulation: 2kV; Height: 170mm; Diameter: 110mm; Weight: 1kg; Hole for thermometer. Operation temperature: 1525 °C; Relative humidity: 25 to 80% Atmosphere pressure: 84106.7 kPa(630800 mmHg) This device was also used in USSR and Russian army. So it has best quality. Longevity. And permanence characteristics. Condition: USED( lightly used. From an old stock) Item may have some visible scraches or bad paint etc. Tested and 100% workable. If necessary. The device can be calibrated and provided with the calibration cer


This is a CPI CO. RESISTOR STANDARD RESISTANCE 0.01 OHM( 0.01Ω) This item is in working condition and being sold as is. See photos for item details. Hawaii. Alaska, PR, will have added shipping surcharges, please ask for quote prior to bidding. Powered by The free listing tool. List your items fast and easy and manage your active items.


This is a RESISTOR STANDARD RESISTANCE P310 0.01 OHM( 0.01Ω) This item is in working condition and being sold as is. See photos for item details. Hawaii. Alaska, PR, will have added shipping surcharges, please ask for quote prior to bidding. Powered by The free listing tool. List your items fast and easy and manage your active items.


For sale lot of 3 pc. 0.001 0.01 0.1 Ohm Ω Lot of 3pc. Resistance Standard Resistor Accuracy 0.01% Technical characteristics: Accuracy class: 0.01% Nominal power: 0.1 Watt Max power: 1 Watt Dimensions 170x110 mm Weight: 0.9 kg If you need just one of coils. Just let me know or see my other items. I ship by air mail with tracking. Paypal Payment. 14 days Refund or Item Change!


